|Minimum Order Quantity||1 Unit|
|Brand||Furuno Electric Co. Ltd.|
|Measuring Parameter||Speed of sound (SOS)|
|Measurement Type||Dry type (acoustic gel used)|
|Dimensions||W525 mm x D310 mm x H200 mm|
|Measuring Site||Calcaneus (Heel Bone)|
|Measurement Precision||%CV 0.5 % or better (in test cases measurement)|
|Operation Humidity||30 to 85% RH|
|Frequency||50 or 60 Hz|
|Measurement Method||Ultrasound Pulse Penetration|
|Result Display||SOS, T-Score, Z-Score, % YAM, % AGE Measurement result with Graphic display|
|External Interface||USB, Bluetooth|
|Measurement Time||Approx. 10 seconds|
|Ultrasound Frequency||Center frequency 500 kHz|
|Operating Condition||Temperature- 10 to 35 Degree C, Humidity- 35 to 85 %RH (non condensing)|
|Supply Voltage||AC 100-120 V, 50 / 60 Hz, 0.6A (max), AC 200-240 V, 50 / 60 Hz, 0.3A (max)|
Bone Densitometer offers you portable and accurate technology for measuring bone properties at the heel without the use of ionizing radiation. When measured at the heel, it helps you accurately predict fracture risk for post-menopausal women. Our CM Series of bone ultrasonometers is built on advanced QUS technology proven of many research studies.
Background: Reduced bone mineral density (BMD) is associated with renal osteodystrophy and osteoporosis in end-stage renal failure patients. Dual-energy X-ray absorptiometry (DXA) is the standard non-invasive method to assess BMD, but is not always widely available. Quantitative heel ultrasound (QUS) is a mobile, relatively inexpensive, easy to perform and radiation-free method which can predict fractures to the same extent as DXA. This study assessed the usefulness of QUSvsDXA in determining BMD in chronic haemodialysis patients.
Methods: Patients had their BMD at the hip and spine measured by DXA. QUS of the left heel (CM-300) measured speed of sound (SOS). Correlations between DXA and QUS parameters were calculated. Receiver operator characteristic (ROC) curves were plotted for SOS and used to define cut-off points for calculating sensitivities and specificities for SOS. Femoral neck BMD was applied as the standard for diagnosing osteoporosis (T≤−2.5) and osteopenia (T>−2.5 and ≤−1) by WHO criteria.
Results: Eighty eight patients (45.5% women), mean age 58±17 years, were studied. A total of 19% and 49% had femoral neck BMDs in the osteoporosis and osteopenia ranges, respectively.
Conclusions: DXA and QUS parameters were significantly correlated. However, sensitivities and specificities of QUS parameters were not sufficiently high for QUS to be used simply as an alternative to DXA. The relatively high negative predictive values suggest that QUS may reliably screen out patients unlikely to have a BMD in the osteoporotic range. The relatively low positive predictive values, however, mean that subjects classified as osteoporotic using QUS require further investigations such as DXA to confirm the diagnosis.